HANNA Ammonia low Range Handheld Colorimeter
The HI-700 Ammonia Low Range Checker is a simple, accurate and cost effective way to measure low ranges of ammonia in fresh water.
Designed as a more accurate alternative to chemical test kits, the HI-700 provides quick, accurate results in four easy steps.
HI-700 uses an adaptation of the ASTM Manual of Water and Environmental Technology, D1426-92, Nessler method. The reaction between ammonia and reagents causes a yellow tint in the sample.
- Freshwater Aquariums
- Water Quality
- Small palm-sized design
- Accurate from ±0.05ppm ± 5% of reading
- Dedicated to a single parameter
- Uses a single AAA battery
- Auto Power Off after 10 minutes in inactivity
- Built-in timer for timed reaction readings
Four steps. One click. Instant Readout.
1. ‘Zero’ the Checker®HC with your vial & 10ml water sample inside
2. Take vial out and add your reagent
3. Place your vial into your Checker®HC
4. Press the button and read the results
Battery powered – Allows use away from a power source. Batteries are included with all battery operated products.
Use Reagents HI-700-25 with the HI-700 Ammonia Low Range Colorimeter - Checker®HC, x 25 tests.
Use Calibration Check Set HI-700-11 with the HI-700 Ammonia Low Range Colorimeter - Checker®HC.
|Range||0.00 to 3.00 ppm NH3-N|
|Accuracy @ 25°C||±0.05 ppm ±5% of reading|
|Light source||LED @ 470 nm|
|Light Detector||Silicon photocell|
|Method||Adaptation of the ASTM Manual of Water and Environmental Technology D1426-92, Nessler Method. The reaction between ammonia and reagents causes a yellow tint in the sample.|
|Environment||0 to 50°C; RH max 95% non-condensing|
|Battery Type||(1) 1.5V AAA|
|Auto-off||After ten minutes of non-use|
|Dimensions||81.5mm x 61mm x 37.5mm|
|Ordering Information||HI-700 Checker® is supplied with (2) sample cuvettes with caps, ammonia reagent starter kit for 25 tests (1 bottle of HI-700A-0 reagent A and 1 bottle of HI-700B-0 reagents B), battery, instructions.|
|PLEASE NOTE:||In some cases, samples may contain other parameters which may interfere with results|
|Interference may be caused by:||Hardness above 1 g/L
Various aliphatic and aromatic amines